
The Radiometer Equation for Position Switched Measurements
Defining the following:
we can calculate
where we have neglected terms quadratic in . Noting that
we see that Equation F.7 and Equation F.6 differ by only a constant (which is the volttotemperature scaling factor). Therefore
If one is taking multiple ON scans per OFF scan, then the time spent to acquire a scan is given by
where N is the number of ON source measurements per OFF source measurement. Therefore, becomes:
To determine the optimum OFF source integration time when one is acquiring multiple ON source measurements per OFF source measurement, we find the minimum of Equation F.10 with respect to:
which reduces to
Table F.1: 2Bit Correlator Efficiencies
where we have used Equation F.3. Inserting this value for into Equation F.10 leads to the following relation for the optimal scan rms where one observes N ON source measurements per OFF source measurement
See Ball (1976, “Methods of Experimental Physics”, volume 12, part C, page 52) for a parallel discussion on noise considerations for position switched measurements. The factor takes account of the efficiency for quantized correlation spectrometers like the 12m MAC. Cooper (1976, “Methods of Experimental Physics”, volume 12, part B, page 284) discusses this correlator efficiency and quotes the values noted in Table F.1. Note that the quoted values assume Nyquist sampling. Many correlators, including the MAC, oversample by some amount, so the correlator efficiency is actually somewhat higher than that for Nyquist sampled data. The theoretical calculations are always predicated on having rectangular passbands of exactly the nominal bandwidth, whereas the effective bandwidth with practical filters is always smaller. This and the fact that there are various nonideal properties of digitizers will tend to reduce the efficiency. Thus, for the purpose of estimating the required observing time to reach a given sensitivity, users should use the theoretical efficiency for Nyquist sampling since the benefit of oversampling will be cancelled by miscellaneous other losses. Formulas for calculating the correlator efficiency for arbitrary oversampling factors can be found in Thompson, Moran, and Swenson (1986) and are listed in Table F.1. For the MAC, which is a 2bit, 3level correlator, the correlator efficiency is plotted in Figure F.1.

Copyright Arizona Radio Observatory. 